CDV+CAV-II+CIV Antigen Combo Rapid Test Cassette (Secretions), Canine Distemper Virus Antigen
|CDV+CAV-II+CIV Antigen Combo Rapid Test Cassette (Secretions), Canine Distemper Virus Antigen
|No Equipment Required
|Easy Visually Interpretation
CDV+CAV-II+CIV Antigen Combo Rapid Test Cassette (Secretions) is a combined cassette to differentially diagnose the presence of Canine Distemper Virus antigen (CDV Ag) and canine adenovirus-II antigen (CAV-II Ag) and canine influenza virus antigen (CIV Ag) in canine’s ocular and nasal secretions .
CDV+CAV-II+CIV Antigen Combo Rapid Test Cassette (Secretions) is based on sandwich lateral flow immunochromatographic assay. The test cassette has three testing windows. The testing window has an invisible T (test) zone and C (control) zone. When sample is applied into the sample well on the cassette, the liquid will laterally flow on the surface of the test strip. If there is enough CDV or CAV-II or CIV Ag in the sample, a visible T line will appear. The C line should always appear after a sample is
applied, indicating a valid result. By this means, the cassette can accurately indicate the presence of CDV or CAV-II or CIV Ag in the sample.
Do not use after expiration date.
• Handle all specimens as if they contain infectious agents. Observe established precautions against microbiological hazards throughout testing and follow the standard procedures for proper disposal of specimens.
• Wear disposable gloves and eye protection when specimens are being tested
• Humidity and temperature can adversely affect results.
• Do not remove test cassette from its pouch until immediately before use.
• Do not reuse the test kit.
• Do not mix components from different lot and different products.
DIRECTIONS FOR USE
Allow the test cassette, specimen, buffer, and/or controls to equilibrate to room temperature (15-30 °C) prior to testing.
1.Collect canine’s ocular and nasal secretions with the swab provided. Make the swab wet sufficiently.
2.Insert the wet swab into the provided buffer tube. Agitate it to assure good sample extraction.
3.Place the test cassette on a clean and level surface. Hold the dropper vertically and transfer 3 drops of extracted sample (approximately 120 μl) to each specimen well (S) of the test cassette, then start the timer. See illustration below.
4.Read the result in 5-10 minutes. Do not interpret results after 15 minutes.
INTERPRETATION OF RESULTS
Positive: The presence of both C line and T line, no matter T line is clear or vague.
Negative: Only clear C line appears.
Invalid: No colored line appears in C zone, no matter whether T line appears.
|CDV+CAV-II+CIV Antigen Combo Rapid Test Cassette